Be sure to review the Instruction Manual for your Power Meter before using the current range (compatible). HIOKI,PW,04,V, ser *3: For detailed specifications of Model , see the comparison chart on page 6. 1. 0 5. Can be read directly by manually setting the CT ratio. DIGITAL POWER HiTESTER , , , AC/DC POWER HiTESTER , , AC/DC DIGITAL HiTESTER , .
|Published (Last):||25 August 2009|
|PDF File Size:||11.19 Mb|
|ePub File Size:||16.9 Mb|
|Price:||Free* [*Free Regsitration Required]|
All reasonable care has been taken in the production of this manual, but if you find any points which are unclear or in error, please contact your supplier or the International Sales and Marketing Department at HIOKI headquarters.
Messages are provided in the interface to control the Power Meter. There are two types of messages: Instruction to set the voltage range. Character data is not case-sensitive, although response messages from the instrument are only upper case.
The output queue is also cleared in the following circumstances: Events are what trigger service requests. Out of Range Bit 6 Over A peak overflow of voltage or current occurred in the active power DataIntegrate integration value.
Bit 5 CurrentOver A peak overflow of current occurred in the current integration DataIntegrate value. Event Status Register 0: Event Status Register 1: Command Execution Time Command execution time indicates the time for analyzing and processing long form commands.
Queries the integration set time and the: Device-specific Commands Frequency Range [Zero-crossing Filter] The frequency range and zero-crossing filter settings are linked. Can be omitted Description Page Response data for queries: Device-specific Commands Measurement Value Output: Data Formats Reference Response data Message [ ]: Can be omitted Description Page for queries: Turns OFF all output items: ALLClear including harmonic wave.
Data by current range during auto-range integration: Data Formats Reference Message [ ]: Can be omitted Description Page Response data for queries Harmonic wave measurement: Queries the RSC setting items. Backup data error Description Perform the instrument self-test and return the result as a numerical value 0 to 4. Returns zero when no error occurs. The Status Byte Register bits corresponding to the event status registers are also cleared.
Although NRf numerical values are accepted, values to the right of the decimal are truncated. Set and Query the Integration Time Command: The setting range is 1 minute to 9, hours and 59 minutes. The unit is 1 minute. Releases the currently held display value. Displays the maximum value maximum value hold. Displays the minimum value minimum value hold. Sets the instrument as the slave device.
Sets the instrument as the master device. The unit is in volts [V]. The numerical value is accepted in NRf format.
Query Returns the voltage range setting in NR1 format. The unit used for current is amperes [A]. Unlike other commands such as: Query Queries zero-cross threshold values.
Providing the finest test equipment solutions since 1992
Parameter List Description Values in parentheses can 33322 be used. See by current range Note 1. Normal measurement values HRMS: Harmonic wave level display, all orders a: Harmonic wave content display, all orders a: ALLClearthe measurement 3332 for the items shown in display areas a through d will be output. SEParator command to change the message unit separator mxnual a semicolon “;” Data by current range See Note 2. Data by current range Integration Instantaneous value Perform and Query a Reset of Output Items: Output Items Measurement status data: Output Items Time Average Current: Query Outputs the items specified via: Output Items Command Syntax: Output Items Order Command Syntax: Odd orders only EVEN: Returns the RSC baud rate and execution confirmation message settings Description as string values.
When set to ON, a response is also received when sending a command.
In addition, an execution confirmation message is included after the response to a query. The Bluetooth adapter is required to be initialized just one time with Model PW Description Although NRf numerical values are accepted, values to the right of the decimal are truncated.
While the display is heldcannot be executed. Status Descriptions Integration Reset Integration calculations are stopped and the integration time and integration value are reset.
Register for items that have caused an error. The program data elements used in expressions are character program data and binary numerical hhioki program data. Also See for PW Instruction manual – pages.
Page of Go.
Advanced search – Hioki 3332
Page 23 – Device-specific Commands Measurement Va Page – Status-dependent Commands Device-specif Communication Command Instruction Manual. This manual explains the communication commands for the above Power Meter. Be sure to review the Instruction Manual for your Power Meter before using the. Please refer to the instruction manual for your Power Meter for details regarding. All reasonable care has been taken in the production of this manual, but if you find any.
In the interest of product development, the contents of this manual are subject. The content of this manual is protected by copyright. No reproduction, duplication or modification of the content is permitted without.
Table of Contents Add to my manuals Add. Summary of Contents for Hioki PW Page 1 All reasonable care has been taken in the production of this manual, but if you find any points which are unclear or in error, please contact your supplier or the International Sales and Marketing Department at HIOKI headquarters.
Hioki Clamp-On Power Logger with 4x A CTs | TechRentals Wiki
Table Of Contents Contents 1 Introduction Command Execution Time Command Execution Time Command execution time indicates the time for analyzing and processing long form commands.
Device-specific Commands frequency Range [zero-crossing Filter] Device-specific Commands Frequency Range [Zero-crossing Filter] The frequency range and zero-crossing filter settings are linked. Page 31 Data Formats Reference Message [ ]: Page 32 Data Formats Reference Message [ ]: Page 61 Parameter List Description Values in parentheses can also be used. Page 62 Parameter List Description Values in parentheses can also be used.
Page 63 Parameter List Description Values in parentheses can also be used.
Page 79 Set and Query: Page 81 Set and Query: Page 82 Set and Query: Page 83 Set and Query: Page 33322 Set and Query: Page 85 Set and Query: Page 86 Set and Query: Page 87 Set and Query: Page 88 Set and Query: Page 89 Set and Query: